Payment & Shipping Terms:
|Spectral Range:||8~14μm||Frame Rate:||1~30Hz|
RoHS Infrared Thermal Camera Module,
Infrared Thermal Camera Module 256x192,
LWIR Uncooled Thermal Camera
Small Size 256x192 / 12μm Uncooled Focal Plane Array LWIR Infrared Thermal Camera Module
TIMO256 is a tiny thermal sensor module with ultra compact, simplifying package design. It integrates optical lens, 256x192 / 12μm wafer level package (WLP) long wave infrared detector and basic image processing circuit to quickly obtain thermal images of the target area and heat distribution.
TIMO256 infrared camera module is fully optimized with small size, light weight & low price (SWaP) to meet the strict requirements of customers in the infrared thermal imaging applications, but it still provides better image quality. Its super miniature structure, compact package dimension and ultralow power consumption are convenient to be integrated into various inspection tools, security monitoring cameras or handheld infrared thermal imagers.
- Focal plan array VOx microbolometer
- 256x192 / 12 µm uncooled long wave infrared (LWIR) detector
- Low power consumption micro thermal camera module
- High precision human body / industrial temperature measurement
|IR Detector Performance|
|Focus Mode||Fixed Zoom|
|Depth of Field||10cm to Infinity|
|Frame Rate||1~30Hz (Customizable)|
|Temperature Range||-20°C ~ +120°C (Customizable)|
|Temperature Accuracy||Customizable (Meet the Requirements of Body or Industrial Thermography)|
|Dimension (mm)||15x13x6.83 (The Specifications Shall Prevail)|
|Operation Temperature||-20°C ~ +60°C|
|Storage Temperature||-40°C ~ +85°C|
The TIMO256 uncooled LWIR thermal module is widely used in many areas, such as Thermography, Intelligent Hardware, Smart Building, Smart Home, AIoT etc.
1. What is NETD?
NETD refers to "Noise Equivalent Temperature Difference". It is a measure for how well a thermal imaging detector is able to distinguish between very small differences in thermal radiation in the image. NETD is typically being expressed in milli-Kelvin (mK). Usually the lower the NETD figure is, the higher the thermal sensitivity indicates.
2. What is SWaP3?
SWaP3 refers to size, weight, power, performance and price. It indicates the technology trend of infrared industry.
3. What is DRI?
DRI usually refers detection, recognition and identification distance of a thermal imaging system. It is a crucial parameter to judge the performance of the thermal imaging system.
Contact Person: Sales Manager